Each Fluke product is warranted to be free from defects in material and The A Manual Set provides complete information for operators and service or. Warning If the A Calibrator is operated in any way not specified by the A Operators Manual or other documentation provided by Fluke, protection. View and Download FLUKE A operator’s manual online. Multi-Product Calibrator. A Test Equipment pdf manual download.
|Published (Last):||23 February 2018|
|PDF File Size:||10.60 Mb|
|ePub File Size:||7.40 Mb|
|Price:||Free* [*Free Regsitration Required]|
This feature is used for ac power, dc power, dual dc volts and dual ac volts outputs. Program the Calibrator Mainframe for an output of 5.
Equipment Required for Calibration and Verification The equipment listed in Table is required to calibrate and verify performance of the A. Remove the fulke Phillips screws from the bottom cover. For the volts-volts phase test, ac couple the input to the phase meter as shown in Figure This byte is cleared set to 0 when the power is turned on.
FLUKE A OPERATOR’S MANUAL Pdf Download.
You can have the A source a dc current below 1. Enter text from picture: Remove the bottom cover. Appendices Glossary scale manuzl Same as gain error.
Return to Rl and flatten the ledge as much as possible. Rms Versus Peak-to-peak Waveforms Figure Follow these steps to calibrate high frequency Feveled Sine Wave flatness for the amplitude being calibrated.
Record the period 5500aa the table and compare to the tolerance column. Page 54 A Operator Manual Now adjust pot Rl, and observe the ledge that occurs within the first 2 ns of the rising edge.
Edge Specifications Table For this test, set the A output to sinewave.
Fluke 5500a – Sc300 Multi-product Calibrator
All equipment specified for SC calibration must be calibrated, certified traceable if traceability is to be maintained, and operating within their normal specified operating environment. Press the numeric keys and decimal point key to enter the desired frequency output maximum five numeric keys.
Figure shows the connections and the equation you use to calculate actual resistance.
When you press a key, the name of the key shows on the display. Adjust the main time 55000a position and vertical offset until the edge signal is centered on the display. The 10 ms to 2 ps markers are derived from a square wave signal that is generated on the A6 board and passed to the A50 board for wave shaping and external trigger generation.
Allow the A reading to stabilize, then record the A rms reading for each wave type and voltage in Table If this warning occurs recheck the setup and carefully re-enter the actual resistance insuring proper multiplier i. Adjusting the Leveled Sine Wave Function. Diagnostics runs the following tests: Equipment Setup for High Lrequency Llatness Test the High current function. Typically this board shows aberrations of 0. Equipment Setup flukr Ftigh Frequency Flatness All high frequency flatness procedures use the following equipment.
This is the first of five calibration menus, which you can scroll through by pressing the softkey under MODE. Once you have turned Rl to the point at which the spur is at a minimum, the signal is balanced between the VCOs, and you have completed the adjustment. Verification at 1 Mf2 1.
Adjust A90R12 to flatten out the edge signal. Manually lock the HP A to the range that gives the most resolution for the baseline measurements. Set the communications parameters to match that of the A. Connect the thermocouple to the front panel TC connector.